Effective treatment using a Tens Unit pads only happens when the electrodes are used with enjoyable, prickling, rubbing type activation. The activation is creating sensors firings of neurological fibers that makes those neurological anxiety engulf the Pain Nerves, which are transferring the agony sensation activation. There is no magic; it is simply a procedure for understanding what is the purpose of placing the Tens Unit pads, generally self-sticking recyclable electrodes, in the appropriate place.
Unlike homeopathy where the hook is little, a Tens Unit pads has a much bigger activation place and uniqueness of placement is not as critical. Small needles in homeopathy often are designed to get rid of a power pain path, whereas electric activation is can be unpleasant and attack the region thusly disturbing and changing the power. The top of an electrode 2 x 2 inches or 4×4 inches wide of place per electrode equating a total of 8 sq. inches wide, where the homeopathy needles surface area is that of the hook itself periods the depth the hook is placed.
Acupuncture is a procedure for finding and placement into a place a needle(s) whereas electric activation (e-stim) is a procedure for hanging over and transferring power in a place. Interferential has a much higher frequency (rate-how many periods machine goes off and on per second). The higher the speed, the less the level of resistance is, and the higher the transmission of the electric activation. Acupuncture needles have to be placed past the point of most level of resistance (skin or scars) to get into the region of power to get rid of the circulation.
The distribution of the power with reusable electrode pads covers a much bigger place and increases the chances of pain activation disruption.
Electrode placement is completely about passing electricity between two or four electrodes through the agonizing place. As the agony sensation goes, which is very common for serious pain conditions such as arthritis, fibromyalgia syndrome, RSD, back problems, radiating sciatic sensors pain, joint pain, foot pain, pain, RLS -Restless Leg Syndrome, shoulder pain, etc., then you can reposition the electrodes to pass the activation through the agony sensation area(s).
Electrodes come in many varied styles and sizes, from butterfly to square to rectangle-shaped to round and many more. The dimensions of the reusable electrode pads chosen are done to protect the region being treated. If the agony sensation emanates in the backbone and the agonizing place is 6 inches wide or so long then frequently a 1″ x 6″ staggered placement or an Oval electrode, electrode will be used, so the 6 inches wide per SSA can be used to totally activate the agonizing backbone place. If the affected person has joint pain then a compact sized 2″ Round electrode may be used to protect the popliteal space behind the joint, which has a compact sized footprint, but the activation covers the joint pain place.
Often the two electrodes per route can be of differing style depending on the harshness of the agony sensation in a specific place. This technique enables the electric activation to be targeted by power, particularly when you are using interferential treatment. Interferential treatment requires bigger electrodes due to what is called “current density”.
Current solidity is comparable to one status in a bath that provides 10 gallons per minute of water through only one little gap. If the affected person were status a few feet away under that polished brass bath go then the single stream of water would be unpleasant due to the pressure behind the circulation. Bathing would be an agonizing procedure. That is get over, by rather than one little gap add 100 more to the polished brass bath go, and now the sensation of showering is comfortable and possible painless. With interferential the same principles apply to electrodes. Reusable electrode pads bigger than typical Tens Unit pads are used to reduce the present solidity by increasing the activation place.